I do not get any results when I read one kinetic point in VICTORDocument ID: DQ056
Relevant to following products: WorkOut 2.0
Document description: Why canít I see any results with a kinetic read of 1 point?
When reading kinetics if you set the repeat count in the VICTOR to 1 then no results will appear when the assay is run. This is because the kinetic views cannot show only 1 point; and it will appear as if no measurements have been taken. The software gives no warning of this situation; if you are reading Kinetics By Wells then it is expecting kinetic measurements (i.e. more than 1 reading per well). If you want to read only 1 point per well then use the VICTOR Label option instead of K.